In addition, their existing systems:
- required expensive “green board” custom hardware
- had proprietary software
- and maintenance was expensive and unreliable
“They wanted to find and implement an alternative solution. They felt they had an opportunity to reduce manufacturing costs by minimizing product giveaway, if they could just meet their check weighing specifications consistently, “ explains Area Business Manager, Stan Modzel. At the same time they wanted to install checkweighers that would:
- use off-the-shelf, open source, non-proprietary technology
- be easy for the operators and maintenance folks to use, modify & maintain
- retrofit into existing system
Hardy’s Engineering Manager, Dean Raby presented the Customer with the solution: “We offered them multiple HI4050CW check weighing bundles. This solution met all their needs.” The check weighing bundle consists of a digital, ceramic load cell and check weighing controller:
- It was easy to install, use, maintain & modify. It is cost effective. It’s not a black box, it is an off-the-shelf solution. It easily integrates to Rockwell and other PLC’s.
- It was able to manage up to 250 packs per minute. It breaks the traditional analog strain gauge “speed barrier”, without the cost of MFR (Magnetic Force Restoration or Other similar high speed / resolution technologies)
- It delivered many smart check weighing features. It offers advanced EDGE DETECTION (weight on & weight off) eliminating the need for photo eyes. It provides the operator with real time indication of DATA QUALITY. Plus it has an embedded DATA LOGGER, which can be used during startup or trouble shooting.
The customer’s operational and engineering teams ran an off-line trial which produced promising results. A decision was made to install 12 x HI4050CW bundles onto one of their production lines. Dean Raby, Hardy’s Engineering Manager monitored and verified the results by doing random checkweighs manually over a period of more than 4 weeks. “All 12 product inspections systems were monitored and upstream filling systems adjusted. The mean difference and standard deviation measured improved over the 4 week period. No safety issues were experienced,” Dean confirms. The baseline on the previous systems was delivering = 0.54g, with a standard deviation = 0.82. The final analysis with the new Hardy HI4050CW bundles delivered a weight variance mean of = 0.11g with a standard deviation = 0.20. The results were better than the 0.25g specification they were trying to attain. No production line stoppages occurred during the test phase and no issues were found with secondary weight checks.
“The customer calculated that the reduction in finished product giveaway would deliver savings of up to 1 Million dollars per month per manufacturing facility. This was based on installing 160 checkweighing bundles on 14 manufacturing lines in their packaging area, and running 3 shifts,” Area Business Manager Stan Modzel explains.